Internal tube 1 - Halogen-quenched, uncompensated GM pancake tube with
thin mica window, 1.4-2.0 mg/cm2 areal density. Effective diameter of window is
45 mm (1.75 in.).
Internal tube 2 - Halogen-quenched, uncompensated thin wall beta-gamma
GM peanut tube. Radiation symbol on front and top labels mark the center of the
μR/h - 0 to 2,000,000 μR/h
mR/h - 0 to 2,000 mR/h
μSv/h - 0 to 20,000 μSv/h
cps - 0 to 12,500 cps
cpm - 0 to 750,000 cpm
Total counts - > 99,999,999,999
Accuracy: ±15% over the entire operating range
Windows Mobile™ PocketPCs, Palm™ Pilots or Windows™ based Laptop or Desktop
System Requirements: Windows XP, 98, 2000, and Windows
Mobile and PalmOS
Outputs: Bluetooth Wireless (30 ft range): count to
computer, handheld or PDA phone.
Gamma Sensitivity: 7.5 CPS/mR/hr referenced to Co60.
Smallest detectable level for I-125 is .02 μCi at contact.
Response Times: Slow = 30 seconds, Medium = 15 seconds, and
Fast = 10 seconds
Alert Set Range: Software Alarm - mR/hr .001 - Full
Range for both Accumulated Dose and Exposure Rate. The Probe’s internal alarm is
Exposure Rate only.
Timer: Total Count timer from 1 -99,999,999,999 seconds.
Count Light: Red LED flashes with each radiation event.
Audio Indicator: Internally mounted beeper (can be switched
off, volume adjusted and can be set for single tube audio).
Anti-Saturation: Meter will hold at full scale in fields as
high as 100 times the maximum reading.
Power Requirements: Two (2)
AA alkaline batteries.
Temperature Range: -10ºC to 50ºC (14ºF to 122ºF).
Weight: 337.4 grams (11.9 oz.) with battery.
Size: 332.7 x 63.5 x 35.6 mm (13.1 x 2.5 x 1.4in.)
90 day Warranty on GM Detector, 1 year limited
Back to Abacus